The Industry Standard in Depth Sounder Test Sets
[ Depth Sounder Test Set ]
[ Chirp Depth Sounder Test Set]
[ Computer Controlled Depth Sounder Test Set]
[ Low Frequency Universal Transducer Test Set] [ Universal Transducer Test Set] [ Multi Sensor Transducer Test Set ]
[ Computer Controlled Transducer Test Set ] [ Loran-C Signal Generator ] [ Downloads]
The DSTS-4A Depth Sounder Test Set uses digital signal processing to bring new levels of accuracy and ease of use together for your benefit. Equally at home in the engineering lab, repair shop, or production line, the DSTS-4A fills all the requirements to test and calibrate today's sophisticated depth sounders.
[Options for the DSTS-4A]
More About the DSTS-4A
By design, the DSTS-4A dramatically shortens the time spent repairing and calibrating sonar systems. Technicians and engineers know how important it is to view all the pertinent data on one screen. A two by twenty-four character super-twisted LCD display gives you complete information on the depth sounder under test. Data on the depth sounder under test is logically presented. The top line of the display shows the incoming pulse parameters; specifically, the frequency, pulse width, pulse period, and the pulse voltage. The second line displays the outgoing pulse parameters, frequency, width, and depth. You always know what you are measuring without reaching for a switch. And, with digital signal processing, what you see is accurate to the last digit.
Depth units may be set for feet, fathoms, or meters. Calibration is based on 4800 feet per second for feet and fathoms, and 1500 meters per second for metric.
Options include an RS-232 port, built-in rechargeable batteries, and a broadband balun for use with depth sounders that have balanced outputs.
Using the DSTS-4A is easy! Connect it to a depth sounder's transducer port. Read the pulse output frequency, width, amplitude, and period. Check the receiver sensitivity by adjusting the reply echo amplitude. You now know the exact condition of the sounder. That's all there is to it! Reply echo frequency tracks the input frequency, so you don't have to bother setting it. The exclusive Auto-width mode tracks the incoming pulse width on a pulse-to-pulse basis. This is very important, as many depth sounders use varying pulse widths for noise rejection and ranging. If the received echo doesn't track the transmitted pulse, the depth sounder may reject it. Auto-width mode insures that the depth sounder gets a realistic copy of its transmitted pulse.
For more sophisticated tests, the echo frequency, depth, and width are variable over the entire range. The Fish Echo feature generates an additional reply echo preceding the bottom echo. The DSTS-4A turns the fish echo on and off for selected numbers of input pulses. This is very useful when evaluating a depth sounders ability to detect fish or small objects above the bottom. Individual toggle switches increment or decrement the outgoing pulse parameters for the bottom and fish echoes. There are no complicated keyboards or arcane procedures to remember.
A scope trigger output occurs ten microseconds before the reply pulse is generated, so you can track the reply echo as it passes through the various stages of the receiver.
The DSTS-4A has three internal load impedances, and an external load position. The external load position connects to the front panel banana jacks. You can now use a custom load without difficulty. The internal loads are parallel resistor-capacitor combinations with the following values:
|Input frequency||2.9 to 999.9 KHz||0.1 KHZ||± .02% ±100 Hz 1|
|Input width||202 to 20,000 uS||1 uS||± .02% ±2 cycles|
|Input period||10 to 9999 mS||1 mS||± .02% ±1 mS|
|Input amplitude||50 to 2500 Vp-p||10 Vp-p||± 5% ±10 Vp-p|
|Min input level||303 Vp-p||N/A||± 10 Vp-p|
|Output frequency||2.9 to 999.9 KHz||0.1 KHZ||± .05%|
|Output width||10 to 20,000 uS||100 uS||± .02%|
|Output depth||1 to 999.9 ft/mt/fa||0.1 unit||± .02%|
|Output depth X 10||1 to 9999 ft/mt/fa||1.0 unit||± .02%|
|Output level||10 to 150,000 uV||1-2-5 steps||± 1.5 dB 4|
|Vprop ft, fm||4800 to 5000 f/s||1 unit||± .02%|
|Vprop mtrs||1460 to 1530 m/s||1 unit||± .02%|
|Power requirements: 10-15 VAC or 12-18 VDC @ 120 mA nominal.|
|Size and Weight: 10" W by 7"D by 4.5"H. Shipping Wt. 6 Lbs.|
BAL-550, Balanced to Unbalanced Line Converter.
Control all DSTS-4A functions such as depth, width, fish echo, frequency, etc. Reply level control option has 0.3 dB resolution and 76 dB range. Read the input pulse voltage, frequency, width, period, and DSTS modes. Commands are simple, "d1347" sets the depth to 134.7 units, "E" turns on the fish echo, "e" turns it off, "I" reads input pulse data, etc.. The serial data format is 9600,8,N,1, and cables are supplied with a DB-9 female connector for direct connection to any PC.
Please note that both the above RS-232 control options function identically with one exception:
The EAB has an electronic attenuator to control the reply echo amplitude, whereas the RS232-BI-DIR does not. Existing DSTS-4A units with RS-232 options are easily retrofitted for reply amplitude control with the EAB/2U or EAB/3U upgrade package.
Rechargeable Battery Pack, BP-4A
The BP-4A will power the DSTS-4A for up to 5 hours of continuous use, and charges automatically whenever the DSTS-4A is connected to an external power source.
Please E-Mail email@example.com or telephone (757) 366-9211 (within the USA) for more information on EDI's product design services.
Click here to email your questions or comments to EDI.
[ Home ]
[ Depth Sounder Test Set ]
[ Options for Depth Sounder Test Set ]
[ Computer Controlled Depth Sounder Test Set ] [ Multi-Sensor Transducer Test Set ]
[ Downloads ]
1708 South Park Court, Chesapeake, VA 23320. Phone (757) 366-9211
Cat DSTS4A/3 01/98